To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
A Nichols radiometer is the apparatus used by Nichols and Hull in 1901 for the measurement of radiation pressure. It consisted of a pair of small silvered glass mirrors suspended in the manner of a torsion balance by a fine quartz fibre within an enclosure in which the air pressure could be regulated. The torsion head to which the fiber was attached could be turned from the outside by means of a magnet. A beam of light was directed first on one mirror and then on the other, and the opposite deflections observed with mirror and scale. By turning the mirror system around so as to receive the light on the unsilvered side, the influence of the air in the enclosure could be ascertained. This influence was found to be a minimum, and to have an almost negligible value, at an air pressure of about 16 mmHg (2.1 kPa). The radiant energy of the incident beam was deduced from its heating effect upon a small blackened silver disk, which was found to be more reliable than the bolometer when it was first used. With this apparatus the experimenters were able to obtain an agreement between observed and computed radiation pressures within about 0.6 %. The original apparatus is at the Smithsonian.
This apparatus is sometimes confused with the Crookes radiometer in which vanes turn in a partial vacuum under the influence of low pressure gas molecules and not directly by the photons themselves.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Nichols_radiometer". A list of authors is available in Wikipedia.|