To use all functions of this page, please activate cookies in your browser.
my.chemeurope.com
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
 My watch list
 My saved searches
 My saved topics
 My newsletter
Electronmultiplying CCD
Additional recommended knowledgeAn electronmultiplying CCD (EMCCD, also known as an L3Vision CCD, L3CCD or Impactron CCD) is a chargecoupled device in which a gain register is placed between the shift register and the output amplifier. The gain register is split up into a large number of stages. In each stage the electrons are multiplied by impact ionization in a similar way to an avalanche diode. The gain probability at every stage of the register is small (P < 2%) but as the number of elements is large (N > 500), the overall gain can be very high (g = (1 + P)^{N}), with single input electrons giving many thousands of output electrons. Reading a signal from a CCD gives a noise background, typically a few electrons. In an EMCCD this noise is superimposed on many thousands of electrons rather than a single electron; the devices thus have negligible readout noise. EMCCDs show a similar sensitivity to Intensified CCDs (ICCDs). However, as with ICCDs, the gain that is applied in the gain register is stochastic and the exact gain that has been applied to a pixel's charge is impossible to know. At high gains (> 30), this uncertainty has the same effect on the signaltonoise ratio (SNR) as halving the quantum efficiency with respect to operation with a gain of unity. However, at very low light levels (where the quantum efficiency is most important) it can be assumed that a pixel either contains an electron  or not. This removes the noise associated with the stochastic multiplication at the cost of counting multiple electrons in the same pixel as a single electron. The dispersion in the gain is shown in the graph on the right. For multiplication registers with many elements and large gains it is well modelled by the equation: if where P is the probability of getting n output electrons given m input electrons and a total mean multiplication register gain of g. Because of the lower costs and the somewhat better resolution EMCCDs are capable of replacing ICCDs in many applications. ICCDs still have the advantage that they can be gated very fast and thus are useful in applications like rangegated imaging. EMCCD cameras indispensable need a cooling system to cool the chip down to temperatures around 170 K. This cooling system unfortunately adds additional costs to the EMCCD imaging system and often yields heavy condensation problems in the application.
Commercial EMCCD cameras typically have clockinduced charge and darkcurrent (dependent on the extent of cooling) that leads to an effective readout noise ranging from 0.01 to 1 electrons per pixel read. Custombuilt deepcooled noninverting mode EMCCD cameras have provided effective readout noise lower than 0.1 electrons per pixel read[1] for lucky imaging observations. See also
Commercial links

This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Electronmultiplying_CCD". A list of authors is available in Wikipedia. 