To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
Thermowells are used in industrial temperature measurement to provide isolation between the temperature sensor and the process environment. They are intrusive fittings and are subject to static and dynamic fluid forces. These forces govern their design. Vortex shedding is the dominant concern as it is capable of forcing the thermowell into flow-induced resonance and consequent fatigue failure. The latter is particularly significant at high fluid velocities.
Additional recommended knowledge
The ASME Performance Test Code (PTC 19.3) is the most widely used basis for thermowell design. It is currently being updated to cover a broader range of thermowell designs and fluid conditions.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Thermowell". A list of authors is available in Wikipedia.|