To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
A Bradbury-Nielsen shutter (or Bradbury-Nielsen gate) is a type of electrical ion gate, which was first proposed in an article by Norris Bradbury and Russel A. Nielsen, where they used it as an electron filter. Today they are used in the field of mass spectrometry where they are used in both TOF mass spectrometers and in ion mobility spectrometers, as well as Hadamard transform mass spectrometers (a variant of TOF-MS). The Bradbury-Nielsen shutter is ideal for injecting short pulses of ions and can be used to improve the mass resolution of TOF instruments by reducing the initial pulse size as compared to other methods of ion injection.
Additional recommended knowledge
Theory of Operation
The concept behind the Bradbury-Nielsen shutter is to apply a high frequency voltage in an 180° out-of-phase manner to alternate wires in a grid which is orthogonal to the path of the ion beam. This results in charged particles only passing directly through the shutter at certain times in the voltage phase (φ=nπ). At other times the ion beam is deflected to some angle by the potential difference between the neighboring wires. This deflection is divergent with ions that pass through alternate slits being deflected in opposite directions.
Micromachined ion gates
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Bradbury-Nielsen_shutter". A list of authors is available in Wikipedia.|