To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
Fluorescence cross-correlation spectroscopy
Additional recommended knowledge
Fluorescence cross-correlation spectroscopy (FCCS) was introduced by Eigen and Rigler in 1994 and extends the fluorescence correlation spectroscopy (FCS) procedure by introducing high sensitivity for distinguishing fluorescent particles which have a similar diffusion coefficient. FCCS uses two species which are independently labelled with two spectrally separated fluorescent probes. These fluorescent probes are excited and detected by two different laser light sources and detectors commonly known as green and red respectively. Both laser light beams are focused into the sample and tuned so that they overlap to form a superimposed confocal observation volume.
The normalized cross-correlation function is defined for two fluorescent species and which are independent green, G and red, R channels as follows:
where differential fluorescent signals at a specific time, and at a delay time, later is correlated with each other.
Cross-correlation curves are modeled according to a slightly more complicated mathematical function than applied in FCS. First of all, the effective superimposed observation volume in which the G and R channels form a single observation volume, in the solution:
where and are radial parameters and and are the axial parameters for the G and R channels respectively.
The diffusion time, for a doubly (G and R) fluorescent species is therefore described as follows:
where is the diffusion coefficient of the doubly fluorescent particle.
The cross-correlation curve generated from diffusing doubly labelled fluorescent particles can be modelled in separate channels as follows:
In the ideal case, the cross-correlation function is proportional to the concentration of the doubly labeled fluorescent complex:
Contrary to FCS, the intercept of the cross-correlation curve does not yield information about the doubly labelled fluorescent particles in solution.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Fluorescence_cross-correlation_spectroscopy". A list of authors is available in Wikipedia.|