To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
Gas chromatography ion detector
Additional recommended knowledge
Principle of Operation
An ion detector is analogous to a capacitor or vacuum tube. It can be envisioned as two metal grids separated by air with inverse charges placed on them. A electric potential difference (voltage) exist between the two grids. After components are ionized in the detector, they enter the region between the two grids, causing current to pass from one to the other. This current is amplified and is the signal generated by the detector. The higher the concentration of the component, the more ions are generated, and the greater the current.
Some early FIDs actually used two metal grids as their ion detectors. However, more efficient designs have been developed, so few current ion-type detectors use two metal grids. But the principle is the same, and it can be easiest to think of the detector in this manner.
Types of Ion Detectors
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Gas_chromatography_ion_detector". A list of authors is available in Wikipedia.|