To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
The Weston cell, invented by Edward Weston in 1893, is a wet-chemical cell that produces a highly stable voltage suitable as a laboratory standard for calibration of voltmeters. It was adopted as the International Standard for EMF in 1911.
Additional recommended knowledge
As shown in the illustration, the cell is set up in an H-shaped glass vessel with the cadmium amalgam in one leg and the pure mercury in the other. Electrical connections to the cadmium amalgam and the mercury are made by platinum wires fused through the lower ends of the legs.
The original design was a saturated cadmium cell producing a convenient 1.0183 Volt reference and had the advantage of having a lower temperature coefficient than the previously used Clark cell. (Reference cells must be applied in such a way that no current is drawn from them.)
The temperature coefficient can be reduced by shifting to an unsaturated design, the predominant type today. However, an unsaturated cell's output decreases by some 80 microvolts per year, which is compensated by periodical calibration against a saturated cell.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Weston_cell". A list of authors is available in Wikipedia.|