To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
In electronics and in electrical engineering, thermocouples are a widely used type of temperature sensor and can also be used as a means to convert thermal potential difference into electric potential difference. They are cheap and interchangeable, have standard connectors, and can measure a wide range of temperatures. The main limitation is accuracy; system errors of less than one degree Celsius (°C) can be difficult to achieve.
Additional recommended knowledge
Principle of operation
In 1821, the German–Estonian physicist Thomas Johann Seebeck discovered that when any conductor (such as a metal) is subjected to a thermal gradient, it will generate a voltage. This is now known as the thermoelectric effect or Seebeck effect. Any attempt to measure this voltage necessarily involves connecting another conductor to the "hot" end. This additional conductor will then also experience the temperature gradient, and develop a voltage of its own which will oppose the original. Fortunately, the magnitude of the effect depends on the metal in use. Using a dissimilar metal to complete the circuit creates a circuit in which the two legs generate different voltages, leaving a small difference voltage available for measurement. That difference increases with temperature, and can typically be between one and seventy microvolts per degree Celsius (µV/°C) for the modern range of available metal combinations. Certain combinations have become popular as industry standards, driven by cost, availability, convenience, melting point, chemical properties, stability, and output. This coupling of two metals gives the thermocouple its name.
It is important to note that thermocouples measure the temperature difference between two points, not absolute temperature. In traditional applications, one of the junctions—the cold junction—was maintained at a known (reference) temperature, while the other end was attached to a probe.
Having available a known temperature cold junction, while useful for laboratory calibrations, is simply not convenient for most directly connected indicating and control instruments. They incorporate into their circuits an artificial cold junction using some other thermally sensitive device, such as a thermistor or diode, to measure the temperature of the input connections at the instrument, with special care being taken to minimize any temperature gradient between terminals. Hence, the voltage from a known cold junction can be simulated, and the appropriate correction applied. This is known as cold junction compensation.
Additionally, a device can perform cold junction compensation by computation. It can translate device voltages to temperatures by either of two methods. It can use values from look-up tables or approximate using polynomial interpolation.
A thermocouple can produce current, which means it can be used to drive some processes directly, without the need for extra circuitry and power sources. For example, the power from a thermocouple can activate a valve when a temperature difference arises. The electric power generated by a thermocouple is a conversion of the heat energy that one must continuously supply to the hot side of the thermocouple to maintain the electric potential. The flow of heat is necessary because the current flowing through the thermocouple tends to cause the hot side to cool down and the cold side to heat up (the Peltier effect).
Thermocouples can be connected in series with each other to form a thermopile, where all the hot junctions are exposed to the higher temperature and all the cold junctions to a lower temperature. Thus, the voltages of the individual thermocouple add up, which allows for a larger voltage and increased power. With the radioactive decay of transuranic elements providing a heat source this arrangement has been used to power spacecraft on missions too far from the Sun to utilize solar power.
Usually the thermocouple is attached to the indicating device by a special wire known as the compensating or extension cable. The terms are specific. Extension cable uses wires of nominally the same conductors as used at the thermocouple itself. These cables are less costly than thermocouple wire, although not cheap, and are usually produced in a convenient form for carrying over long distances—typically as flexible insulated wiring or multicore cables. They are usually specified for accuracy over a more restricted temperature range than the thermocouple wires. They are recommended for best accuracy.
Compensating cables on the other hand, are less precise, but cheaper. They use quite different, relatively low cost alloy conductor materials whose net thermoelectric coefficients are similar to those of the thermocouple in question over a limited range of temperatures, but which do not match them quite as faithfully as extension cables. The combination develops similar outputs to those of the thermocouple, but the operating temperature range of the compensating cable is restricted to keep the mis-match errors acceptably small.
The extension cable or compensating cable must be selected to match the thermocouple, meaning that the conducting alloys must match. Usage of an extension or compensation cable of a different alloy will introduce a new thermocouple junction, distorting the readings. The cables generate a voltage proportional to the difference between the hot junction and cold junction, and are connected in the correct polarity so that the additional voltage is added to the thermocouple voltage, compensating for the temperature difference between the hot and cold junctions.
The relationship between the temperature difference and the output voltage of a thermocouple is nonlinear and is approximated by polynomial:
The coefficients an are given for n from zero to between five and nine.
To achieve accurate measurements the equation is usually implemented in a digital controller or stored in a look-up table. Some older devices use analog filters.
A variety of thermocouples are available, suitable for different measuring applications. They are usually selected based on the temperature range and sensitivity needed. Thermocouples with low sensitivities (B, R, and S types) have correspondingly lower resolutions. Other selection criteria include the inertness of the thermocouple material, and whether or not it is magnetic. The thermocouple types are listed below with the positive electrode first, followed by the negative electrode.
Type K (chromel–alumel) is the most commonly used general purpose thermocouple. It is inexpensive and, owing to its popularity, available in a wide variety of probes. They are available in the −200 °C to +1200 °C range. The type K was specified at a time when metallurgy was less advanced than it is today and, consequently, characteristics vary considerably between examples. Another potential problem arises in some situations since one of the constituent metals, nickel, is magnetic. The characteristic of the thermocouple undergoes a step change when a magnetic material reaches its Curie point. This occurs for this thermocouple at 354°C. Sensitivity is approximately 41 µV/°C.
Type J (iron–constantan) is less popular than type K due to its limited range (−40 to +750 °C). The main application is with old equipment that cannot accept modern thermocouples. J types cannot be used above 760 °C as an abrupt magnetic transformation causes permanent decalibration. The magnetic properties also prevent use in some applications. Type J thermocouples have a sensitivity of about 50 µV/°C.
Type N (nicrosil–nisil) thermocouples are suitable for use at high temperatures, exceeding 1200 °C, due to their stability and ability to resist high temperature oxidation. Sensitivity is about 39 µV/°C at 900°C, slightly lower than type K. Designed to be an improved type K, it is becoming more popular.
B, R, and S
Types B, R, and S thermocouples use platinum or a platinum–rhodium alloy for each conductor. These are among the most stable thermocouples, but have lower sensitivity, approximately 10 µV/°C, than other types. The high cost of these thermocouple types makes them unsuitable for general use. Generally, type B, R, and S thermocouples are used only for high temperature measurements.
Type B thermocouples use a platinum–rhodium alloy for each conductor. One conductor contains 30% rhodium while the other conductor contains 6% rhodium. These thermocouples are suited for use at up to 1800 °C. Type B thermocouples produce the same output at 0 °C and 42 °C, limiting their use below about 50 °C.
Type R thermocouples use a platinum–rhodium alloy containing 13% rhodium for one conductor and pure platinum for the other conductor. Type R thermocouples are used up to 1600 °C.
Type S thermocouples use a platinum–rhodium alloy containing 10% rhodium for one conductor and pure platinum for the other conductor. Like type R, type S thermocouples are used up to 1600 °C. In particular, type S is used as the standard of calibration for the melting point of gold (1064.43 °C).
Type T (copper–constantan) thermocouples are suited for measurements in the −200 to 350 °C range. Often used as a differential measurement since only copper wire touches the probes. As both conductors are non-magnetic, type T thermocouples are a popular choice for applications such as electrical generators which contain strong magnetic fields. Type T thermocouples have a sensitivity of about 43 µV/°C.
Type C (tungsten 5% rhenium – tungsten 26% rhenium) thermocouples are suited for measurements in the 0 °C to 2320 °C range. This thermocouple is well-suited for vacuum furnaces at extremely high temperatures and must never be used in the presence of oxygen at temperatures above 260 °C.
Type M thermocouples use a nickel alloy for each wire. The positive wire contains 18% molybdenum while the negative wire contains 0.8% cobalt. These thermocouples are used in the vacuum furnaces for the same reasons as with type C. Upper temperature is limited to 1400 °C. Though it is a less common type of thermocouple, look-up tables to correlate temperature to EMF (milli-volt output) are available.
The table below describes properties of several different thermocouple types. Within the tolerance columns, T represents the temperature of the hot junction, in degrees Celsius. For example, a thermocouple with a tolerance of ±0.0025×T would have a tolerance of ±2.5 °C at 1000 °C.
Thermocouples are most suitable for measuring over a large temperature range, up to 1800 °C. They are less suitable for applications where smaller temperature differences need to be measured with high accuracy, for example the range 0–100 °C with 0.1 °C accuracy. For such applications, thermistors and resistance temperature detectors are more suitable.
Type B, S, R and K thermocouples are used extensively in the steel and iron industries to monitor temperatures and chemistry throughout the steel making process. Disposable, immersible, type S thermocouples are regularly used in the electric arc furnace process to accurately measure the steel's temperature before tapping. The cooling curve of a small steel sample can be analyzed and used to estimate the carbon content of molten steel.
Heating appliance safety
Many gas-fed heating appliances such as ovens and water heaters make use of a pilot light to ignite the main gas burner as required. If the pilot light becomes extinguished for any reason, there is the potential for un-combusted gas to be released into the surrounding area, thereby creating both risk of fire and a health hazard. To prevent such a danger, some appliances use a thermocouple as a fail-safe control to sense when the pilot light is burning. The tip of the thermocouple is placed in the pilot flame. The resultant voltage, typically around 20 mV, operates the gas supply valve responsible for feeding the pilot. So long as the pilot flame remains lit, the thermocouple remains hot and holds the pilot gas valve open. If the pilot light goes out, the temperature will fall along with a corresponding drop in voltage across the thermocouple leads, removing power from the valve. The valve closes, shutting off the gas and halting this unsafe condition.
Some systems, known as millivolt control systems, extend this concept to the main gas valve as well. Not only does the voltage created by the pilot thermocouple activate the pilot gas valve, it is also routed through a thermostat to power the main gas valve as well. Here, a larger voltage is needed than in a pilot flame safety system described above, and a thermopile is used rather than a single thermocouple. Such a system requires no external source of electricity for its operation and so can operate during a power failure, provided all the related system components allow for this. Note that this excludes common forced air furnaces because external power is required to operate the blower motor, but this feature is especially useful for un-powered convection heaters.
A similar gas shut-off safety mechanism using a thermocouple is sometimes employed to ensure that the main burner ignites within a certain time period, shutting off the main burner gas supply valve should that not happen.
Out of concern for energy wasted by the standing pilot, designers of many newer appliances have switched to an electronically controlled pilot-less ignition, also called intermittent ignition. With no standing pilot flame, there is no risk of gas buildup should the flame go out, so these appliances do not need thermocouple-based safety pilot safety switches. As these designs lose the benefit of operation without a continuous source of electricity, standing pilots are still used in some appliances.
Thermopile radiation sensors
Thermopiles are used for measuring the intensity of incident radiation, typically visible or infrared light, which heats the hot junctions, while the cold junctions are on a heat sink. It is possible to measure radiative intensities of only a few μW/cm2 with commercially available thermopile sensors. For example, laser power meters are based on such sensors.
Radioisotope thermoelectric generators
Thermopiles can also be applied to generate electricity in radioisotope thermoelectric generators.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Thermocouple". A list of authors is available in Wikipedia.|