Accurate and Easy with HX204
Moisture control is an important aspect for the production of high quality plastic parts. Measuring moisture content with the METTLER TOLEDO HX204 Halogen Moisture Analyzer is a practical and precise procedure. This application note illustrates the accurate determination of moisture content in plastics with the METTLER TOLEDO HX204 Moisture Analyzer. Using PA 6 as an example, see how easy and fast it is to get moisture results with the HX204 moisture analyzer. In addition, this application note shows how the results match those from the Karl Fischer titration method.
You may also be interested in these white papers
Measuring Moisture in Lithium-Ion Batteries
This application note describes how METTLER TOLEDO halogen moisture analyzers help ensure consistent quality in LiBs with fast and accurate moisture c
How to clean your moisture analyzer?
A clean halogen moisture analyzer is the first step towards safe and accurate moisture content measurements.
Quality Management in Moisture Analysis
The GMAP white paper provides detailed explanations of the critical aspects of moisture analysis quality management
Moisture analysis in chemical industry
What is the best way to determine dry content in the chemical industry?
Drying Oven Method vs. Halogen Moisture Analyzer
This white paper provides practical guidance of how halogen moisture analysis can replace the loss on drying (LOD) using a drying oven method
Guide to moisture analysis
This guide to moisture analysis assists you in determining reliably the moisture content with a halogen moisture analyzer.
A Fast Routine Test – Ensures Trusted Moisture Results
Regular verification of measuring devices is essential to ensure valid results and maintain quality
Data Management with a Halogen Moisture Analyzer
Our reference paper describes various options for data management with a moisture analyzer.