To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
Electrochemical scanning tunneling microscope
The electrochemical scanning tunneling microscope, or ESTM, was invented in 1988 by Kingo Itaya in Japan. With ESTM, the structures of surfaces and electrochemical reactions in solid-liquid interfaces can be observed at atomic or molecular scales.
Additional recommended knowledge
Electrochemical reactions occur in electrolytic solutions—for example electroplating, etching, batteries, and so on. On the electrode surface, many atoms, molecules, and ions adsorb and affect the reactions. In the past, in order to obtain information about the structure of electrode surfaces and reactions, the sample electrode was taken out of the electrolytic solution and measured under ultra high vacuum (UHV) conditions. In this case, the structure of the surface changed and could not be observed precisely. By using this microscope, however, these problems are resolved.
In electrolytic solutions, a very complicated electrical double layer of H2O molecules and anions is formed. In this layer, as the distribution of anions changes with the potential of the electrode, it is necessary to control the reaction on the electrode. The potentials of the working electrodes (the sample and the tip) are controlled independently against a reference electrode. In this case, the tunneling bias voltage is the difference between the two potentials. A count electrode is used in order to flow the current by the reaction on the electrode surface. By using these four electrodes, the electrochemical reaction is controlled precisely by the external voltage, and the surface in liquid can be observed.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Electrochemical_scanning_tunneling_microscope". A list of authors is available in Wikipedia.|