To use all functions of this page, please activate cookies in your browser.
With an accout for my.chemeurope.com you can always see everything at a glance – and you can configure your own website and individual newsletter.
- My watch list
- My saved searches
- My saved topics
- My newsletter
Magnetic force microscope
A magnetic force microscope (MFM) is a scanning probe microscope (SPM) that can map the spatial distribution of magnetism by measuring the magnetic interaction between a sample and a tip. As magnetic devices have become smaller and smaller, an evaluation technique with nanoscale spatial resolution has become necessary. To meet this need, the MFM was developed.
Additional recommended knowledge
The principle of MFM measurement is based on noncontact atomic force microscopy (NC-AFM). Unlike NC-AFM, magnetic materials are used for the sample and tip, so that not only the atomic force but also the magnetic interaction are detected. Many kinds of magnetic interactions are measured by MFM, including magnetic dipolar interaction.
In MFM measurements, the magnetic force between the sample and tip is given by:
Because the magnetic stray field from the sample will affect the magnetized state and vice versa, in most cases it is difficult to obtain quantitative information from the MFM measurement. To interpret the information quantitatively, the configuration of the tip must be known. With this measurement, a typical resolution of 30 nm can be achieved (Abelmann, 1998). The best resolution obtained so far is 18 nm (SwissProbe, August 2005).
A potential method of increasing the resolution would involve using an electromagnet on the tip instead of a permanent magnet. By only enabling the magnetic tip when placed over the pixel being sampled could increase the resolution.
|This article is licensed under the GNU Free Documentation License. It uses material from the Wikipedia article "Magnetic_force_microscope". A list of authors is available in Wikipedia.|